When ventriculoatrial shunt gets lost
- Received: , Accepted: ,
A 40-year-old male, with a medical history of subarachnoid hemorrhage (SAH) from aneurysm rupture complicated by chronic hydrocephalus treated with the ventriculoperitoneal shunt (VPS), presented to the emergency room for recurrence of Hakim–Adams syndrome for the past 2 months. Three years ago, he underwent revision surgery with implantation of a ventriculoatrial shunt (VAS).
The patient was afebrile and clinical examination showed no focal neurologic deficits and no meningismus. The computed tomography (CT) scan of the head revealed a recurrence of chronic hydrocephalus [Figure 1]. The cervicothoracic CT scan revealed that the catheter was disconnected from the valve and had migrated through the right heart chambers in the pulmonary arteries. The patient denied chest pain and dyspnea. Cardiac auscultation was normal. Blood tests revealed no sign of infection (WBC 8.6G/L, CRP 0.9 mg/L) and there was no sign of pulmonary embolism (D-dimer 0.36 µg/mL).
One day after admission, the patient was taken to the interventional radiology suite. A 7F sheath (Radifocus Terumo®, Somerset, NJ 08873 USA) was placed through right femoral venous access. A 5F pigtail catheter (Radifocus Terumo®) was introduced through a 0.035 guidewire (Terumo®) until the catheterization of the right heart: this allowed us to reach the pulmonary arteries. The migrated silicone shunt was stuck, looping in the right inferior subsegmental artery. This loop was gradually undone so that the extremity of the shunt was set free in the inferior vena cava. The pigtail catheter was removed, and a 6F EN snare® lasso retrieval device (Merit Medical Inc., South Jordan, Utah 84095 USA) was used to catch and remove the migrated shunt [Figures 2 and 3].
Two weeks after admission, the patient underwent revision surgery with removal of the remaining proximal shunt, and implantation of a contralateral frontal VPS shunt with an adjustable pressure valve. The patient’s neurological status improved remarkably within days following the surgery. His post-operative course was unremarkable. He was discharged three weeks after admission.
SAH is the second most common etiology of chronic hydrocephalus (17.4%). VPS is the treatment of choice for chronic hydrocephalus; however, this procedure has one of the highest rates of complications in neurosurgery (11–47%), including excessive CSF drainage, shunt obstruction, infection, and abdominal complications. VAS can be performed as a revision surgery or as a first-line treatment for chronic hydrocephalus, depending on the surgeon’s experience or in the case of prior abdominal surgery. VAS is also associated with a 43–50% rate of post-operative complications, including excessive drainage, infection, autoimmune glomerulonephritis, and pulmonary embolism; however, less shunt obstructions are reported compared to VPS.
VAS is a medical device inserted in the superior vena cava through the internal jugular vein, using the Seldinger technique. Thrombus formation at its distal extremity is a rare complication. Migration of the distal catheter in the right chambers of the heart or into the pulmonary arteries is an even rarer complication; it may be life-threatening and has been reported essentially in children, but also in adults, in case of broken distal catheter.[4-6]
Our case report discusses the migration of an intact distal catheter, disconnected from the valve 3 years after its implantation. Unless there are cardiac or pulmonary injuries due to the migration of the catheter, the latter should be urgently extracted by the endovascular procedure, and surgery should be avoided.[4-6] Minimally invasive endovascular retrieval of lost foreign material was a fantasy 50 years ago, but it has rapidly become a preferred solution with the technical improvement of endovascular catheters and biplane imaging.[7,8]
Concerning the VAS, we would recommend revision surgery of the whole system. What is more, to avoid disconnection and migration of the distal catheter, we also recommend implanting shunts with a monobloc valve-distal shunt system or securing the junction between the valve and the catheter with a tight suture. Ideally, the ventricular catheter is placed in the frontal horn of the lateral ventricle so that the valve lies directly on the convexity of the skull and not behind the ear, where repeated neck movements may lead to shunt disconnection. This concept applies to both VAS and VPS.
Neurosurgeons and interventional radiologists are used to working together, for example, in patients with aneurysm rupture. This collaboration is beneficial for an interdisciplinary approach in case of unexpected complications, such as endovascular VAS catheter migration. Indeed, minimally invasive techniques should always be preferred for the extraction of migrated endovascular material in modern medicine.
AcknowledgmentsWe thank Dr. Antoine RIMBOT, from the interventional radiology service of Sainte-Anne Military Teaching Hospital in Toulon, for the clinical care of the patient.
Declaration of patient consentThe authors certify that they have obtained all appropriate patient consent forms.
Financial support and sponsorshipNil.
Conflicts of interestThere are no conflicts of interest.
- J Pediatr Neurosci. 2018;13:78-80Endovascular retrieval of detached ventriculoatrial shunt into pulmonary artery in pediatric patient: Case report.
- [Google Scholar]
- No Shinkei Geka. 2012;40:539-45Intracardiac migration of a ventriculoatrial shunt catheter treated by endovascular transvenous retrieval.
- [Google Scholar]